We have calculated preliminary throughput curves for NIRES. They are posted on the throughput and sensitivity pages.
On 03/05/18 we identified the shadow of a loose screw projected on top a spectral flat. Since a loose metal piece so close to the main detector poised a serious risk, we decided to warm up the instrument. On previous runs, we had also identified the shadows of some other transparent loose pieces that would move away after rotating the instrument.
On 03/17/18 K. Matthews removed the detector from the instrument and found a connector bracket loose in the cavity between the detector and a saphire window. We also found a piece of mylar tape and a couple of washers. After re-attaching the bracket, we proceeded to start pumping the instrument.
On 03/19/18 the instrument was installed back on the telescope
On 02/01/18 Rui Xue and Jingzhe Ma performed the first sceince observations with NIRES.
NIRES leaving Caltech.
NIRES arriving to Mauna Kea.